80T-IR – read user manual online or download in PDF format. Pages in total: Find great deals for Fluke 80t-ir Temperature Probe 80TIR. Shop with confidence on eBay!. Find great deals for Fluke Infrared Temperature Probe M# 80t-ir. Shop with confidence on eBay!.
|Published (Last):||12 June 2007|
|PDF File Size:||13.48 Mb|
|ePub File Size:||15.57 Mb|
|Price:||Free* [*Free Regsitration Required]|
Refer to Figure 1 for equipment configuration. Save to parts list Save to parts list. In between measurements move the UUT to an environment not affected by the heat source.
Sending feedback, please wait The product detailed below complies with the specifications published by RS Components. You have chosen to save the following item to a parts list:. The product does not contain any of the restricted substances in concentrations and applications banned by the Directive, and for components, the product is capable of being worked on at the higher temperatures flukw by lead—free soldering.
Fluke 80T-IR User Manual
The Manufacturers reserve the right to change this Information at any time without notice. Please select an existing parts list. Set the DMM to the 3. Thank you for your feedback.
Fluke 80t-ir Infrared Temperature Probe 80tk Thermocouple Module
May substitute less accurate blackbody if using in conjunction with transfer standard of equivalent accuracy, i. Repeat steps 80t-jr through 5 for the remaining verification points called out in Table 2. With the UUT powered on, check that the battery voltage is at least 7. Save this item to a new parts list. If your instrument fails to pass any performance specifications, return it to the Eluke Eactory or Service Center for repair. Please enter a message.
fuke Search the history of over billion web pages on the Internet. Verification Points Model Blackbody Temperature at 0. Power up the UUT and allow it to stabilize for five minutes. The foregoing information relates to product sold on, or after, the date shown below.
The Manufacturers disclaim all warranties including implied warranties of merchantability or fitness for a particular purpose and are not liable 80t-it any damages arising from your use of or your inability to use the Information downloaded from this website. Record the reading displayed on the DMM.
Save to an existing parts 80t-it Save to a new parts list. Replace the battery if necessary. Recommended Equipment Table 1 lists the equipment required to perform the calibration verification procedure.
Fluke 80T-IR User Manual – Page 1 of 10 |
Place the UUT aperature three inches from the center and perpendicular to both axes of the Blackbody plate. Accuracy figures are valid for a period of one year.
Adjust the setpoint of the Blackbody to output each of the temperatures in Table 2 fluks 0. Reactivate by switching the UUT off and on again. Add to a parts list. You agree that the Information as provided here through an intermediary may not be error-free, accurate or up-to-date and that it is not advice.